Use a good quality USB cable with a length less than 1.5 meters.
Yes, temperature stabilization is required for best results. All devices are calibrated at 22°C.
Refer to the manual for instructions. (provide link to manual)
Use active cooling to stabilize the temperature of the detector. All devices are calibrated at 22°C.
Yes, python and C++ API are available in the downloads section.
DAC values are factory set and should not be changed. The threshold can only be changed for MPX3 devices.
Advacam uses direct conversion technology which eliminates the need for scintillators.